What is the time required to switch between Pulse IV (4225PMU) and CV (4210ACVU) measurements using the 4225-RPM for the 4200A-SCS?
The RPM eliminates the need to re-cable and increases switching speed between Pulsed IV and CV measurements. However, we do not specify the switching time for the RPMs in the data sheets.
A simple test using a MDO3102 yeided the below approximate switch times:
CVU to PMU measurements through RPM = 460ms.
PMU to CVU measurements through RPM = 1700ms.
This FAQ Applies to:
Product Series: Keithley 4200A-SCS Parameter Analyzer
FAQ ID : 469646View all FAQs »