When I use a custom made arbitrary wave form, set the modulation to burst and the trigger source to external it is not possible to get any signal out of the device.
This was a known limitation on AFG1062's that had firmware v1.0.4 or lower. This limitation has been resolved with the installation of firmware v1.0.5 which can be downloaded from Tek.com.
Cette FAQ concerne :
Série du produit : Générateur de Fonctions Arbitraires AFG1000
Numéro de la FAQ 247536Afficher toutes les FAQ »